The poster highlights some of the many innovations implemented in this unique instrument: Minimized volumes save cycle time and contaminated surface. An optimized wash procedure for most effective and quick cleaning. An injection mechanism that protects hydrophilic analytes from the harms of pressure fluctuations. And waste liquid management that keeps your lab bench clean and tidy!
A typical setup as seen in our previous application note employs three pumps – one for loading the sample out of the autosampler’s sample loop, one for diluting the sample’s organic solvent with water for trapping, and one for the analytical gradient, eluting the concentrated analytes off the trapping column onto an analytical column for separation.
We optimized this design a little, replacing the loading and dilution pumps by a single pump. See how it works in our new animation (turn on sound and/or subtitles):
Amyloid peptides (Abeta; Aβ) result from cleavage of amyloid precursor protein (APP) by beta and gamma secretase.The 42-amino acid variant, amyloid β peptide 1–42 is a widely accepted key biomarker for Alzheimer’s disease (AD) together with the total tau (T-tau) and phosphorylated tau (P-tau) protein. Diagnostic accuracy of the Aβ 1–42 /Aβ 1–40 ratio in cerebrospinal fluid (CSF)
compared to the concentration of Aβ 1–42 alone was found to be even more significant.
Currently these peptides are analyzed routinely in CSF by immunoassays, but in recent years, liquid chromatography tandem mass spectrometry (LC-MS/MS) has been investigated for the quantification of Aβ1-38, Aβ1-40, and Aβ1–42.
[Blennow 2010, Lewczuk 2015, Lame 2011]
Our neighbour CRO, Swiss BioQuant, use the Zirconium pump with a new scheme for loading and analyzing samples at the same time using the same binary pump. Dead times of the MS are greatly reduced, so they save a lot of instrument time while achieving exceptional sensitivity and precision.
Prolab Instruments has released a new software interface for its Zirconium Nano-UHPLC pump series for use in chromatography data systems from various vendors. This was requested by many users, and adds to the flexibility of LC system integration. Continue reading Zirconium Software Integrations for OpenLab, Analyst, Chromeleon, HyStar
In our long history of Micro- and Nano-HPLC product developments, we had the chance to collect experience using many approaches on how to fit a thin glass tube into a large (1/16″) stainless steel port, which is what many HPLC and UHPLC systems employ to this day.
Welcome to Prolab Instruments’ new website! The update was long overdue, and we hope the new site is more informative, up to date and friendly to use than the previous one. In case of any rough edges or missing information, we would be happy to hear from you.